Reversible Hydrogen Electrode (RHE) is the reversible electrode kit, in which the of strong acid then stored. It makes the complicated and dangerous set-up of the hydrogen cylinder and avoids the large volume of hydrogen gas exhaustion, even though it has a good potential stability.
The RHE has a large application field, it can be used in a strong acid solution (<pH 2), such as 0.5 mol/L sulfuric acid, also in a neutral solution, when it is applied with a double junction chamber.
Double junction chamber kit
|Catalog No.||Description||Qty||Click for detailed picture.|
|013376||Double junction chamber||1|
|013377||PTFE Cap for double junction chamber||1|
|002222||Platinum counter electrode 5.7 cm||1|
Reversible hydrogen electrode kit
|013598||RHE Reversible hydrogen electrode||1|
|013378||RHEK accessories kit||1|
|Silicone tube (10 cm)||3|
|Female lure fitting||1|
|Length||Approximately 109 mm|
|Diameter||Head: 14 mm
Neck: 6 mm
|Temperature range||5 deg C to 60 deg C|
(using with Double junction chamber)
|0 to 2
0 to 9
Double junction chamber specifications
|Diameter||Head: 12 mm
Neck: 6 mm
Set up of the electrode
RHEK Reversible hydrogen electrode kit only
|Set the RHE Reversible hydrogen electrode and the counter electrode to the electrochemical cell, which contains the strong acid solution.
"Disposable syringe" is attached to the open end of "Silicone tube". After confirming whether "Tubing clamp" is released, pull the plunger of the syringe to fill the solution in the electrode.
Please check if the liquid level in Silicone tube reaches to the white arrow and there is not any air bubbles in the electrode by looking before closing the tubing clamp.
|Close the tubing clamp, after checking that there is no air bubbles in the electrode. And then, the syringe is detached from the silicone tube.|
|Set the potentiostat next to the electrochemical cell. Connect the electrode lead for working electrode (green covering) to the RHE. Please make sure that the leads for reference (white covering) and counter electrodes (red covering) are connected together.|
|Electrolysis is done by amperometric i-t technique or chronoamperometric technique during the charge value gets to about two coulombs. In this case, please make sure setting the minus (-) applied potential based on counter electrode. When you use 1.2 mol/L hydrochloric acid as electrolyte solution, the applied potential is set as -3 V and the period of electrolysis would be about 5 minutes.
When you use 0.5 mol/L of sulfuric acid, -3 V of applied voltage, the electrolysis time would be about 9 minutes.
|The electrochemical test is done by connecting the reference electrode lead to the contact pin of RHE.|
RHEK with Double junction chamber kit
|Fill the chamber with a strong acid, such as HCl, HClO
, H SO , etc.. The RHE would be similar to Standard Hydrogen Electrode (SHE) by using 1.2 mol/L Hydrochloric acid as electrolyte solution.|
Make sure that there is not air bubbles in the solution, to eliminate the bubbles tap gently the chamber with the finger.
|Fit the "Double junction chamber cap" on the chamber with a counter electrode and hold it by something like test tube stand or voltammetry cell. Do not touch the bottom of the tip of the chamber during the preparation.|
|Set the RHE, which was previous preparing with a syringe and tubing clamp setting (see the picture at the left), into the PTFE cap hole. Check if the electrode liquid junction is immersed to the solution.|
|Loose the clamp and then pull the plunger of the syringe to take acid solution in the electrode while pushing the electrode down in the chamber securely.
Please confirm the condition whether the solution reaches the position marked by white arrow and there is no air bubbles in the solution.
|Make sure closing tubing clamp. Then, set your potentiostat next to the electrochemical cell. Connect the electrode lead for working electrode (green covering clip) with the contact pin of your RHE. Please make sure connecting your electrode leads for reference (white covering clip) and counter electrodes (red covering clip) with the contact pin of a counter electrode together.|
|Electrolysis is done by amperometric i-t technique or chronoamperometric technique during the charge value gets to about two coulombs. In this case, please make sure setting the minus applied potential based on counter electrode. When you use 1.2 mol/L hydrochloric acid (HCl) as electrolyte solution, The applied potential is set as -3 V and the period of electrolysis would be about five minutes.
|Replace the test tube stand with your electrochemical cell containing test sample. Place the working electrode and counter electrode. And then, connect among your potentiostat and the electrodes to measure your sample.|